We have developed a semi-analytic model of collapse-mode CMUT . The modelled collapse-voltage is 59 V. Simulated impulse response predicts center frequency of 12.9 MHz and 16.6 MHz at a bias voltage of 100 V and 160 V, respectively. Experimental validation with a manufactured CMUT prototype  shows that the dynamic response and frequency-tunability are modelled with a satisfactory accuracy as shown in Fig. 1.
Further experimental studies show that the center frequency of a collapse-mode CMUT can be controlled between 8 MHz and 15 MHz, if the lower bias voltage range is extended and the driving pulse is optimized. Mechanically-scanned B-mode imaging is performed on a phantom at 8 MHz, 11 MHz, and 15 MHz as a first proof of principle of a frequency-tunable imaging with a collapse-mode CMUT .
A 2x2 mm2 32-element phased-array CMUT is integrated with front-end electronics in a rigid probe prototype and connected to Verasonics system (Kirkland, WA, USA) for 2-D real-time imaging and data acquisition . CMUT imaging performance is quantified in terms of resolution and penetration depth at a range of bias voltages, driving pulse frequencies, and number of pulse cycles. Based on this characterization settings for high-penetration, generic, and high-resolution imaging modes are identified. The first 2-D imaging results are shown.
The developed probe prototype is tested ex vivo in a passive heart platform . Images of an aortic valve acquired in high penetration (6 MHz), generic (12 MHz), and high-resolution (18 MHz) mode combine satisfying image quality and penetration depth between 2.5 cm and 10 cm as shown in Fig. 2.
Next, the CMUT probe prototype is further miniaturized into a 12-Fr steerable, forward-looking ICE catheter. The ICE catheter prototype is tested in vivo using a porcine animal model . Images of an aortic valve are acquired in the three imaging modes with the ICE catheter placed in an ascending aorta at multiple depths. It was found that the combination of the forward-looking design and frequency tuning capability allows visualizing intracardiac structures of various sizes at different distances relative to the catheter tip, providing both wide overviews and detailed close-ups.
 Martin Pekař, Stephan H. M. van Nispen, Rob H. B. Fey, Sergei Shulepov, Nenad Mihajlović, Henk Nijmeijer, Sensors and Actuators A: Physical (in review).
 Dirksen, P. and Lugt, A. van der (2009), Production of Pre-Collapsed Capacitive Micro-Machined Ultrasonic Transducers and Applications Thereof, Patent WO 2009037655 (A2), March 2009.
 Martin Pekař, Wendy U. Dittmer, Nenad Mihajlović, Gijs van Soest, Nico de Jong, Frequency Tuning of Collapse-Mode Capacitive Micromachined Ultrasonic Transducer, Ultrasonics, vol. 74, p. 144-152, 2017.
 Pekař, M., Mihajlović, N., Belt, H., Kolen, A. F., Rens, J. van, Budzelaar, F., Jacobs, B., Bosch, J. G., Vos, H. J., Steen, A. F. W. van der and Rem-Bronneberg, D. (2016) ‘Frequency-Agility of Collapse-Mode 1-D CMUT Array’, in Proceedings - IEEE Ultrasonics Symposium. Tours: IEEE.
 Martin Pekař, Alexander F. Kolen, Harm Belt, Frank van Heesch, Nenad Mihajlović, Imo E. Hoefer, Tamas Szili-Török, Hendrik J. Vos, Johan G. Bosch, Gijs van Soest, Antonius F. W. van der Steen. Preclinical Testing of Frequency-Tunable Capacitive Micromachined Ultrasonic Transducer Probe Prototypes, Ultrasound in Medicine and Biology (in press).
|Titel||Proceedings of the 16th International Workshop on Micromachined Ultrasonic Transducers|
|Status||Gepubliceerd - jun 2017|
|Evenement||16th International Workshop on Micromachined Ultrasonic Transducers (MUT 2017), June 28-29, 2017, Glasgow, UK - University of Strathclyde, Glasgow, Verenigd Koninkrijk|
Duur: 28 jun 2017 → 29 jun 2017
|Workshop||16th International Workshop on Micromachined Ultrasonic Transducers (MUT 2017), June 28-29, 2017, Glasgow, UK|
|Periode||28/06/17 → 29/06/17|